2260 Investigator - Sound Intensity Software
Sound Intensity Software for 2260 Investigator
Building acoustics applications such as reduction indices and leakage detection benefit enormously from the intensity technique. 2260-E Investigator with Intensity Probe Kit form a hand-held solution for sound intensity measurements.
The Extended High Frequency Correction for a probe with ½” microphones and 12 mm spacer has not been working correct in the BZ7205 Sound Intensity software for 2260. Fortunately, the error has now been discovered and corrected with the update of the software to version 2.1.1.
For noise source identification applications, this error has minimal influence. However, for sound power determination, depending on the form of the noise source spectrum, the error could lead to an under estimation of the sound power. As an example, for a broad-band measurement using 1/3-octaves in the frequency range 63 Hz to 6.3 kHz, a sound source with pink noise spectrum will be underestimated by 0.4 dB Linear, and by 0.6 dB A-weighted. For a white noise spectrum the underestimations are 1.1 dB Linear and 1.1 dB A-weighted.
Note: If you have old data sets of pressure or intensity spectra measured with earlier versions of the 2260 program, the Extended High Frequency Correction can be applied using the values in the PDF document “Correction.pdf”.
If you use Noise Explorer 7815, then old data sets will automatically be corrected by Noise Explorer version 4.17.4 or higher.
|correction.pdf||40 kb||Inglese||02 May 2013|
|bz 7205 sound intensity software version 2.1.1.zip||.zip||2631 kb||Inglese||02 May 2013|