High Sensitivity Charge Accelerometer
Type 4383 is a piezoelectric accelerometer with side connector, designed specifically for general purpose, high sensitivity vibration measurements.Angebot anfordern
High Sensitivity Charge Accelerometer Type 4383 is a piezoelectric, DeltaShear®, Unigain® accelerometer with side connector. The accelerometer features 10 – 32 UNF receptacle for output connection, and can be mounted on the object by means of a 10 – 32 UNF threaded steel stud.
- General purpose
- High sensitivity
- Low-level, low-frequency measurement
- Vibration testing and analysis
Type 4384 may be treated as a charge source, and it's sensitivity is expressed in terms of charge per unit acceleration (pC/g).
The DeltaShear design involves three piezoelectric elements and three masses arranged in a triangular configuration around a triangular centre post. The ring pre-stresses the piezoelectric elements to give a high degree of linearity. The charge is collected between the housing and the clamping ring.
The piezoelectric element used is a PZ 23 lead zirconate titanate element. The housing material is titanium.
The sensitivity given in the calibration chart has been measured at 159.2 Hz and an acceleration of 10 g. For 99.9% confidence level, the accuracy of the factory calibration is ± 2%.
- Freq. range
- 0.1 – 8400 Hz
- -74 – 250 ℃
- 17 gram
- 3.16 pC/ms^-2
- Residual Noise Level in Spec Freq Range (rms) ±
- 0.06 mg
- Maximum Operational Level (peak)
- 2000 g
- 10-32 UNF
- Accessory Included
- AO 0038
- Clip/Stud/Screw included
- YQ 2960
- General Purpose, Vibration Test
- Resonance frequency
- 28 kHz
- Maximum Shock Level (± peak)
- 5000 g